We are using the device Keithley and yes i have two different temperatures i differentiated one with hot water. All drivers are installed, and it is shown in MAX. It often occur at some intervals and intervals doesn’t seem continous. I have attached a picture of the current scan setup to the loop. Message 24 of
|Date Added:||25 July 2010|
|File Size:||19.66 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Message 30 of Got the user manual this morning and am reading so I can send commands to the machine.
Instrument Control (GPIB, Serial, VISA, IVI)
Never mind I found it. Message 5 of 5.
Overview This example program illustrates how to separate the data read from a Keithley 27XX device into separate columns on a delimited spreadsheet.
Thank you of the help. Message 29 of Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Message 7 of Click on the Read tab and press execute I meithley not familiar with the express vi “Write to Measurement File”, but you could separate yours data like this: Return Status should be x0.
Continuously Write to Spreadsheet Keithley 27XX – Discussion Forums – National Instruments
Seems you are also confused about the device Keithley each channel as a column in output? I’ve been struggling with this for a while now.
I seem to have a different problem now. I put the GPIB In the capture you can see also the test of the VISA communications. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
keithley driver problems – Discussion Forums – National Instruments
I figured the problem is 27sx the converter. How often does it occur and what steps do you take to reproduce this? It would be helpful if someone can help us resolve this.
Additional Images or Video. Hi DaleYes ,its very odd for me too. If you have the NI If I am reading everything correctly, it looks like the output is just an interlaced 1D array. Message 4 of A frustration that occurs with people using Keithley keithldy in LabVIEW is that, when reading from multiple channels on a Keithley device, all of the data collected from different channels is written into a single 1D array.
Most Active Hardware Boards: Ensure you can talk using the diagnostic tool Message 25 of Can you please look in to it. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Page gives the rate commands. Replace “Read Data Buffer Subset.
In order to resolve this issue, a “Decimate 1D Array” function is used in order to separate values from individual channels into separate rows or columns in an array.