Message 1 of A request doesn’t guarantee that they will develop the driver. For example if the time column on the front panel is a vector array from t0 – t10, there will also be a corresponding sepoint column with s0-s Message 4 of Here, the flow rates of each channel are collected in an array and the whole array is the input for the graph. Message 6 of I hope that makes sense.
|Date Added:||3 August 2009|
|File Size:||64.51 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
I also have the mks C mass controller.
MGC C MKS Intruments Computer Communication – Discussion Forums – National Instruments
I can change these settings on the unit, but I do not know witch settings are appropriate. I labvieww sincerely appreaciate someones help. If it is a commercially available instrument, you can officially request one at: Message 6 of Thanks for the help Raj, -Mike.
Message 6 of 9.
Are you able to communicate with this device using simple serial commands from Measurement and Automation Explorer? Search for basic rs tutorials to gain familiarity with the protocol parity, data bits, etc.
Temporal, have you been able to develop drivers for the MKS flow controller? Once I try to write something like baudrate or word bits into the instrument, it still shows 6647c error. Did you found a driver?
If you are unable to communicate with the device using basic commands through HyperTerminal, and we have verified that the serial ports themselves work fine through a loopback test. You can develop the driver yourself. Message 9 of 9.
MKS Instruments C Multi Gas Controller – Instrument Driver – National Instruments
Most Active Hardware Boards: I’m not able to communicate vie RS, I’ve got no reply and of couse the instrument does not accept the commands. Message 4 of Most Active Hardware Boards: The same flowrate simultaneoulsy recorded in Excel or any other data format which I can analyse later once my experiment labivew finished. Most Active Software Boards: Hi Pushkar, nice to hear you are able to talk to the device.
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. I am new in the world of LabView and electronic communications.
Message 7 of 9. Message 8 of I just want to figure out how I can write data into instrument since.
Message 5 of Or is there an easier way to do it? By the looks of it, I acuired the same driver set you did but cannot communicate with the device! This leads us only to the possibility that the instrument itself is not functional.
Instrument Control (GPIB, Serial, VISA, IVI)
We also have a syndicate of third party vendors that specialize in National Instruments’ products and services. Some of the vendors specialize in driver developme nt.
If you would like to try developing your own instrument driver or modify the existing onewe have documentation, model instrument drivers, and driver templates to help at: I’ve done a simple “loop-back” testto confirm my RS port is working, which it appears to be. Message labvoew of